I’m guessing the changing of the waveform shape causes the dropout. When writing an instrument control application for a serial instrument, you communicate with a COM port rather than the serial instrument connected to the COM port. Most function generators have a built-in dc offset that can be added to any waveform. I’d like to have the change be instantaneous. I have two example driver sets:
|Date Added:||19 July 2005|
|File Size:||46.42 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
However, it is still showing twice the set voltage. The mark LabWindows is used under a license from Microsoft Corporation.
This tool automatically detects USB hardware connected to your host PC and adds a labviee representation to the window. Instument is the A.
The procedure for detecting your device depends on the bus you are using. Procedure Locate the communication ports on the back of the Agilent function generator instrument.
The Keysight A 10 MHz function generator offers uncompromising performance at a lower price for basic functions and waveforms. Can the VISA driver be fixed easily?
I’m guessing the changing of the waveform shape causes the dropout. What is the fuse rating on the main power supply assembly in the A? I’m using function generator A but it is giving twice the output voltage. The recommended calibration interval for these devices is 12 months. You can find these instrument drivers in the Instrument Driver Network.
LabVIEW driver for the 33210A
Here is what the Configure Standard Waveform block of that one looks like by the way. If the waveform labfiew large and you are using RS interface, binary data format is not supported.
It will be tough for me to adjust this one. Connect and Configure Agilent Function Generator Instrument You can connect your Agilent series function generator to your host PC using a method that works for your instrument. Back to Top 2. For this example, focus on the section between the Configure Arbitrary Waveform node and the Enable Output node. Worked like a charm!
There are only two ways to tell somebody thanks: When using the generator manually, I can turn the knob and change the output labvieww real time without delays. Maybe the RC time decay formula? Thanks last modified by shaobing Accepted by topic author jto2.
LabVIEW driver for the A | Keysight (formerly Agilent’s Electronic Measurement)
Why is laview output voltage twice the value I set? How do I sweep in both directions. Consider PXI Arbitrary Function Generators NI PXI Waveform Generators can produce precise waveforms including sine, square, triangle, and ramp as well as arbitrary, user-defined waveforms by using data sequences or by streaming continuously from a host or peer-to-peer instrument within the PXI system. When I increase the amplitude, my function generator switches ranges and I hear a click. Don’t forget to quote your name, company and 33210 number, so we can quickly reach you, if there are any questions from our side!
Conduct Sequencing and Database Reporting With TestStand Test Management Software TestStand, a ready-to-run test management software, is designed to help you develop, execute, and deploy automated test and validation systems faster. Thanks for the suggestion Mateus, but I am trying to automate this labciew as best I can. How do I disable this? Lwbview I published my code All you really need is a sequence structure with the wait inside, pass the error cluster through the sequence structure to make sure you wait the 1 second at the right time.